## volumetric flowrate

in gas chromatography
https://doi.org/10.1351/goldbook.V06649
The flowrate of the mobile phase ($$\text{cm}^{3}\ \text{min}^{-1}$$), normally specified at the column temperature and outlet pressure, although the measurement may be made at ambient temperature and must be corrected accordingly (and possibly also for water vapour present in the flowmeter).
Source:
Orange Book, 2nd ed., p. 102 [Terms] [Book]